<!--
 * @Author: cgl
 * @Date: 2023-04-18 15:12:26
 * @LastEditors: cgl
 * @LastEditTime: 2023-09-04 15:12:05
 * @FilePath: \public-health-emergencies-fed:\新放射\新radiation-hygiene-fe\src\views\person\itemControl\item09.vue
 * @Description: 
 * 
 * Copyright (c) 2023 by cgl/微盛信息产业有限公司, All Rights Reserved. 
-->
<template>
	<div>
		
		<table border="1" cellspacing="0" cellpadding="0" align="center">
			<tr class="title">
				<td rowspan="2">序号</td>
				<td rowspan="2" width="15%">检测项目</td>
				<td rowspan="2">检测方法及条件</td>
				<td >验收检测</td>
				<td >状态检测</td>
				<td colspan="2">稳定性检测</td>
			</tr>

			<tr class="title">
				<td>判定标准</td>
				<td>判定标准</td>
				<td>判定标准</td>
				<td>周期</td>
			</tr>

			<tr>
				<td >1</td>
				<td >胸壁侧射野与影像接收器一致性</td>
				<td >测量胸壁侧射野与台边的距离</td>
				<td >超出台边，<br>并≤5.0 mm</td>
				<td >超出台边，<br>并≤5.0 mm</td>
				<td >超出台边，<br>并≤5.0 mm</td>
				<td >六个月</td>
			</tr>
			<tr>
				<td >2</td>
				<td >光野与照射野一致性</td>
				<td >其他三边</td>
				<td >±5.0 mm 内</td>
				<td >——</td>
				<td >±5.0 mm 内</td>
				<td >六个月</td>
			</tr>
			<tr>
				<td >3</td>
				<td >管电压指示的偏离</td>
				<td >25 kV～32 kV 选 3 个档</td>
				<td >±1.0 kV 内</td>
				<td >±1.0kV内</td>
				<td >±1.0kV内</td>
				<td >六个月</td>
			</tr>
			<tr>
				<td >4</td>
				<td >半值层</td>
				<td >28kV</td>
				<td  class="viewfile" @click="viewInTable('F.2')">查看</td>
				<td >——</td>
				<td >——</td>
				<td >——</td>
			</tr>
			<tr>
				<td >5</td>
				<td >输出量重复性</td>
				<td >28kV</td>
				<td >≤5.0%</td>
				<td >≤5.0%</td>
				<td >——</td>
				<td >——</td>
			</tr>
			<tr>
				<td rowspan="2">6</td>
				<td rowspan="2">特定辐射输出量</td>
				<td >28kV，1m处，Mo/Mo</td>
				<td >＞35.0 μGy/mAs</td>
				<td >＞30.0 μGy/mAs</td>
				<td >——</td>
				<td >——</td>
			</tr>
			<tr>
				<td >28kV，1m处，其他靶/滤过</td>
				<td >建立基线值</td>
				<td >＞70.0% <sup>a</sup></td>
				<td >——</td>
				<td >——</td>
			</tr>
			<tr>
				<td >7</td>
				<td >自动曝光控制重复性</td>
				<td >4cmPMMA</td>
				<td >±5.0%内</td>
				<td >±10.0%内</td>
				<td >±10.0%内</td>
				<td >六个月</td>
			</tr>
			<tr>
				<td rowspan="3">8</td>
				<td rowspan="3">乳腺平均剂量/mGy</td>
				<td >普通模式，4cmPMMA</td>
				<td >＜2.0</td>
				<td >＜2.0</td>
				<td >＜2.0</td>
				<td >六个月</td>
			</tr>
			<tr>
				<td >DBT模式，4cmPMMA</td>
				<td >＜2.0</td>
				<td >＜2.0</td>
				<td >＜2.0</td>
				<td >六个月</td>
			</tr>
			<tr>
				<td >普通模式+DBT模式，<br>4cmPMMA</td>
				<td >＜3.5</td>
				<td >＜3.5</td>
				<td >＜3.5</td>
				<td >六个月</td>
			</tr>
			<tr>
				<td colspan="7">
					<div style="text-align: left; margin-left: 20px;">
						<sup>a</sup> 与基线值比较。
					</div>
				</td>
			</tr>

		</table>

		<el-dialog
			title="不同靶/滤过组合的半值层要求"
			:visible.sync="inTableVisible"
			:close-on-click-modal="false"
			class="standarddialog"
			append-to-body
			v-dialogDrag="true"
		>
			<table class="custable" border="1" cellspacing="0" cellpadding="0" align="center">
				<tr>
					<td width="30%">管电压</td>
					<td width="40%">靶/滤过</td>
					<td width="30%">半值层<br>mmAL</td>
				</tr>
				<tr>
					<td rowspan="10">28kV</td>
					<td>Mo/Mo</td>
					<td>0.30≤HVL≤0.40</td>
				</tr>

				<tr>
					<td>Mo/Rh</td>
					<td>0.30≤HVL≤0.47</td>
				</tr>
				<tr>
					<td>Mo/Cu</td>
					<td>HVL≥0.30</td>
				</tr>
				<tr>
					<td>Rh/Rh</td>
					<td>0.30≤HVL≤0.50</td>
				</tr>
				<tr>
					<td>Rh/Al</td>
					<td>HVL≥0.30</td>
				</tr>
				<tr>
					<td>Rh/Cu</td>
					<td>HVL≥0.30</td>
				</tr>
				<tr>
					<td>Rh/Ag</td>
					<td>HVL≥0.30</td>
				</tr>
				<tr>
					<td>W/Rh</td>
					<td>0.30≤HVL≤0.58</td>
				</tr>
				<tr>
					<td>W/Al</td>
					<td>0.30≤HVL≤0.53</td>
				</tr>
				<tr>
					<td>W/Ag</td>
					<td>0.30≤HVL≤0.60</td>
				</tr>
			</table>
			<div slot="footer">
				<el-button @click="inTableVisible = false">关 闭</el-button>
			</div>
		</el-dialog>
	</div>
</template>

<script>
export default {
  name: "item40",
  data() {
    return {
			inTableVisible: false,
    };
  },
  created() {
  },
  methods: {
		viewInTable(type) {
			this.inTableVisible = true;
		}
  }
};
</script>

<style lang="scss" scoped>
@import './content.scss';
</style>